Testing transistors and diodes with an analogue multimeter
- how to test transistors and diodes with a simple multimeter.
Multimeter Tutorial Includes:
Analogue multimeter DMM digital multimeter Basic instructions on how to use a multimeter Multimeter tutorial and video Making a multimeter voltage measurement How to measure current with a multimeter Making a multimeter resistance measurement Simple diode and transistor test Testing and fault finding a transistor circuit
While many digital multimeters these days have a specific capability for testing diodes and sometimes transistors, not all do, especially the older analogue multimeters that are still in widespread use. However it is still quite easy to perform a simple go / no-go test using the simplest of equipment.
This form of testing is able to detect whether transistor or a diode is operational, and although it cannot provide details of the parameters, this is seldom a problem because these components will have been tested at manufacture and it is comparatively rare for the performance to fall to a point where they do not operate in a circuit. Most failures are catastrophic, rendering the component completely inoperable. These simple multimeter tests are able to detect these problems very quickly and easily.
Diode test with an analogue multimeter
The basic diode test is very simple to perform. Just two tests are needed with the multimeter to ensure that the diode works satisfactorily:
... the band on the diode package represents the cathode....
Step by step instructions:
- In step 3 above the actual reading will depend upon a number of factors. The main thing is that the meter deflects, possibly to half way or more. The variation depends on many items including the battery in the meter, and the range used. The main point to note is that the meter deflects significantly.
- When checked in the reverse direction, silicon diodes are unlikely to show any meter deflection whatsoever. Germanium ones that have a much higher level of reverse leakage current may easily show a small deflection if the meter is set to a high ohms range.
This simple analogue multimeter test of a diode is very useful because it gives a very quick indication of whether the diode is basically operational. It cannot, however, test more complicated parameters such as the reverse breakdown, etc.
Transistor test using an analogue multimeter
The diode test using an analogue multimeter can be extended to give a simple and straightforward confidence check for bipolar transistors. Again the test using a multimeter only provides a confidence check that the device has not blown, but it is still very useful.
The test relies on the fact that a transistor can be considered to comprise of two back to back diodes, and by performing the diode test between the base and collector and the base and emitter of the transistor using an analogue multimeter, the basic integrity of the transistor can be ascertained.
It should be noted that a transistor cannot be functionally replicated using two separate diodes because the operation of the transistor depends upon the base which is the junction of the two diodes, being one physical layer, and also very thin.
Step by step instructions:
The instructions are given primarily for an NPN transistor as these are the most common types in use. The variations are shown for PNP varieties - these are indicated in brackets (.. .. ..):
- The final checks from collector to emitter ensure that the base has not been "blown through". It is sometimes possible that there is still a diode present between collector and base and the emitter and the base, but the collector and emitter are shorted together.
- As with the germanium diode, the reverse readings for germanium transistors will not be as good as for silicon transistors. A small level of current is allowable as this results from the presence of minority carriers in the germanium.
Analogue multimeter overview
While most of the multimeters on sale today are digital ones, there are nevertheless many analogue meters still in use. While they may not be the latest in technology, they are still ideal for many uses, and can be used easily for measurements such as the ones above.
Although the tests detailed above are aimed at analogue meters, similar ones can be undertaken with digital multimeters (DMMs)